So you’ve scoped your job, you’ve chosen your equipment, and you’ve landed on the ever-versatile rotary laser. So what’s next? How do I use it?
In this blog, we’re going to show you the best tips, tricks, uses, and a step-by-step guide on how to set up and use your new rotary laser!
So let’s cover the basics..
What is a Rotary Laser?
A rotary laser (also known as a rotary laser level) is an advanced laser level that spins a beam of lightning-fast enough to give the optical illusion of a solid 360-degree plane. These planes can be horizontal, or vertical. Illuminating not just a solid and static line, but lateral planes.
Not to be confused with a laser ‘measuring tool’ which defines space and distance between multiple objects, or a laser pointer which is used in presentations, or playing for endless hours with your cat; a laser level sends a bright beam of vertical light in the form of a line onto flat a surface with supreme accuracy.
For example, a rotary laser level is helpful whether being used inside or outside. It’s used to determine and continuously show a consistent level of elevation when building up, driving, and/or drilling down.
Most rotary lasers will also self-level – typically when positioned within four degrees of level (depending on the manufacturer). Some will also let the user know if it has been thrown off-course. It’s often referred to as a ‘height alert’ feature – meaning the laser will stop working and alert you if it’s height has been bumped out of level.
Step-by-Step: How to Use a Rotary Laser
A manual level can do the job, but for large construction projects; it’s simply not efficient enough. This is where rotary laser levels become the users’ best friend.
So, take it out of the box, and let’s get started!
Step 1. How to Calibrate a Rotary Laser Level
- Place your laser level on a flat surface or tripod in a room with two walls that oppose each other.
- Situate the unit 2.5cm from one wall, and at least 50cm from the other wall.
- Align the laser along the axis of the vial (bubble) you want to level. The vial should point to both walls.
- There are two (2) screws on your device that will correspond to each vial. Tune these screws until the device is level along the two axes. Both vials should show bubbles resting between the two black lines.
- Turn on the laser
- Visually mark where the laser hits the closest and furthest walls.
- Reposition the laser 2.5cm from the furthest wall.
- Level the laser and adjust the height (if using a tripod) allowing your beam to correspond with the mark you made on the other wall.
- Circle back to the first wall and measure the difference between the first mark and the laser line.
Your rotary laser is calibrated correctly if the difference is less than or equal to your level’s specified accuracy
- If it is larger, your rotary laser needs more calibration (continue)
- Adjust the laser so it beams in the middle of the two marks you measured (this mark is true level)
- Lastly, remove the screw cap for the bubble vial and adjust the vial until it is level.
Congratulations, your unit is now correctly calibrated. Time to get to work!
Step 2. How to Use a Rotary Laser Level
Set up your rotary laser level. This type of level is best used on an elevated flat surface. A tripod is the best choice but sawhorses, buckets, and step ladders will also suffice.
- PLACE the level on your chosen surface and turn it on
- If the device is self-leveling, allow it time to find level (one-two minutes).
- If it requires manual leveling, consult the user’s guide on the proper procedure to ensure correct measurements.
- LISTEN for the receiver to beep to indicate a connection with the laser.
- The laser cannot be seen in broad daylight.
- AIM the rotary laser level at the surface at which you want to find a level.
- Make marks and lines to guide installation.
- REPEAT this step for new lines, planes, and levels that need to be found.
Remember that rotary lasers come with many different battery options. Some laser levels will allow you to connect to a wired socket. Whereas others will be relying on battery packs whilst out on site. So don’t forget to TURN OFF your laser when you have made your markings.
For All Your Rotary Laser Needs Pick Position Partners
Position Partners has the tools, equipment, and rotary lasers you need to get your project leveled and underway. Give the team at Position Partners a call today on 1300 867 266 to find out more about how a quality approved and powerful rotary laser tool will benefit your business and future projects.
Don’t forget to explore the full range of Position Partners Lasers and Levels within our online products.